1. Water spray system, good spray effect
2. Prevent back absorption of absorbent
3. Built-in glass filler
4. The use of high vacuum ball grinding mouth has good air tightness
After the circulation pump is turned on, the absorption liquid is pumped into the shower from the round-bottomed flask via the condensing line through the circulation pump, sprayed from the tower cap down through the absorption tower and re-entered into the absorption tank, contact with the absorption liquid filled with glass filler, so that the gas and liquid are fully mixed, so as to achieve the purpose of absorption. If the temperature of the exhaust gas is high, the condensing pipe needs to be connected to cool the absorption liquid. This device can be carried out with an active absorption device, such as using nitrogen as a carrier gas, or vacuum to guide through the absorption device.
The scrubbing liquid contained in the scrubber needs to be selected according to the nature of the gas to be purified and impurities. For acidic impurities, alkaline detergents (such as sodium hydroxide solution) are usually used; for alkaline impurities, acidic detergents (such as chromic acid lotion) can be used; For sexual impurities, you can use oxidizing detergents (such as potassium permanganate solution), etc. When choosing a detergent (a certain solution), you should ensure the following two points:
① The solute in the detergent can chemically react with the impurity gas, so that the impurity can be converted into precipitation or become soluble substances to dissolve. Alternatively, the solvent in the detergent can fully dissolve the impurity gas, thereby achieving the purpose of removing the impurity gas.
②The main gas has a low solubility in the detergent and is not absorbed by the detergent in a large amount. For example, when the chlorine containing hydrogen chloride is washed with saturated brine, a large amount of hydrogen chloride is dissolved in the brine, and the chlorine is not easily dissolved in the saturated brine, thereby achieving the purpose of removing the impurity gas. For another example, when using concentrated sodium bicarbonate to wash carbon dioxide containing hydrogen chloride, hydrogen chloride reacts with sodium bicarbonate to be fully absorbed, carbon dioxide does not react with sodium bicarbonate, and has low solubility in the solution, thereby achieving the purpose of removing impurity gases. If sodium carbonate solution is used instead, sodium bicarbonate is generated by the reaction of carbon dioxide and sodium carbonate solution, and the carbon dioxide is also fully absorbed by the solution. This method is not advisable.
We have a standard production workshop and a complete production team. They always produce a serious and responsible attitude to produce each product. Our R&D team is made up of multi-aspects professional engineers such as design engineer,technology specialist ,qualified specialist and process control manager, etc. Our qualified specialists strictly realize the process comprehensive supervision based on our quality process control system, which contains products design, research and development, test, production, quality inspection and packaging.
Our laboratory instruments and equipment are used in many scientific laboratories, scientific research institutions, universities, hospitals, centers for disease control and other institutions, and are combined with many well-known biopharmaceuticals, new energy, electronics companies, food processing, environmental protection, etc. Enterprise cooperation, our stable and reliable products are exported to more than 120 countries around the world, and have been highly recognized by them, believe LABOAO, we grow together!